Please note: using the search filters may not provide accurate search results for records prior to November 2021. When searching for older records, please ensure that you use the include archive checkbox.
Displaying 1 - 4 of 4 items.
Fetal Spiral Electrode
Philips has found that the spiral tip of the Fetal Sprial Electrode (FSE) may break off during use, potentially requiring surgical intervention to remove the broken tip from the patient. Based on Philips investigation, this can occur due to over rotation…
RecallHealth product recall | 2022-12-15
The pagewriter tc30/50/70 cardiograph device could potentially fail to charge while plugged into the ac power outlet. This is due to a faulty mainboard. based on a supplier analysis, the faulty mainboard was caused by…
RecallHealth product recall | 2022-05-30
The esophageal/rectal/skin temperature probe's instructions for use, (IFU), contain inadequate instructions related to the cleaning and disinfection process for the reusable probes. The IFU instructs the user to clean the product with disinfecting agents…
RecallHealth product recall | 2022-02-21
St80i Stress Test System
When patient demographic information is entered into the system and the patient information is saved or exported to a folder, in very rare situations, the demographic information that is displayed is that of the previous patient instead of the current…
RecallHealth product recall | 2022-01-31