Contact the manufacturer if you require additional information.
|Lot or Serial number
|Model or Catalogue number
|Pagewriter Tc70 Electrocardiograph
|Pagewriter Tc50 - Electrocardiograph
The pagewriter tc30/50/70 cardiograph device could potentially fail to charge while plugged into the ac power outlet. This is due to a faulty mainboard. based on a supplier analysis, the faulty mainboard was caused by delamination of an ic chip after mold assembly.
Recall start date: May 19, 2022
Philips Medical Systems
3000 Minuteman Road, Andover, Massachusetts, Unites States, 01810