Returning search results with filters:
Remove filter forRecall class: Type I
Remove filter forType: Recall Clear all
Please note: using the search filters may not provide accurate search results for records prior to November 2021. When searching for older records, please ensure that you use the include archive checkbox.
Displaying 1 - 15 of 323 items.
Surgical Gut - Plain and Absorbable Sutures
Medtronic was made aware of a supplier calibration issue during the contract sterilization process that may lead to unsterile product. Recall start date: March 1, 2023
RecallHealth product recall | 2023-03-17
CAYSTON: Vials may be cracked in affected lots
Vials may be cracked in affected lots.
RecallHealth product recall | 2023-03-03
Infinity Occipitocervical Upper Thoracic System Multi Axial Screw
It was identified by Medtronic that an incorrect type of titanium was used for the manufacturing of subcomponents of multi-axial screws that are for use as part of the Infinity oct system. This incorrect material reduces the clamping strength of the…
RecallHealth product recall | 2023-03-01
Tri-Staple™ 2.0 Black Intelligent Reload
Specific lots have the potential for a broken sled vane component. A broken sled vane may cause the reload to misfire leading to non-functional staple line closure, transecting tissue without forming staples, and tissue hang-up. These conditions may…
RecallHealth product recall | 2023-02-08
Minicap Disconnect Cap With Povidone-Iodine
Baxter Corportation is issuing a recall for certain lots of the Minicap. These devices are packaged in foil pouches, which may have been incorrectly sealed, i.e., the pouches may have open or weak seals. Recall start date: February 01, 2023
RecallHealth product recall | 2023-02-08
APO-AMITRIPTYLINE 10mg: NDMA Impurity
Affected lot exceeds concentration limit for N-nitrosodimethylamine (NDMA) at the 48-month stability testing timepoint.
RecallHealth product recall | 2023-02-07
Novum IQ Large Volume Pump
Through ongoing internal testing, Baxter Corporation has observed isolated incident of over infusion on the Novum IQ LVP infusion pump during test runs at 0.1 ml/hr after 40 hours of continuous infusion within the 96-hour set change interval. Baxter…
RecallHealth product recall | 2023-02-06
Novastar Full Face Mask
In case the magnets of the Novastar TS and Novastar Plus are positioned near medical devices (implanted as well as not-implanted) and metallic implants the magnetic field might impact the implants or devices and cause serious injury. Recall start date…
RecallHealth product recall | 2023-01-24
Belzer MPS and Belzer UW: Particulate matter
Presence of particulate matter in affected lot.
RecallHealth product recall | 2023-01-18
Medline Custom Procedure Packs - Non-absorbable sutures
Medline Industries, LP is recalling custom procedure packs containing medtronic non-absorbable sutures due to a manufacturing error in the packaging of the sutures that may cause a gap or wrinkle in the breather pouch seal resulting in a sterile barrier…
RecallHealth product recall | 2023-01-11
Nuclear Medicine Systems
For certain nuclear medicine systems, GE Healthcare has become aware that a mitigation may not be correctly implemented. If this is the case, the detector can fall leading to life-threatening bodily injury. Recall start date: December 18, 2022
RecallHealth product recall | 2022-12-30
The replacement silicone sound abatement foam installed into the Trilogy 100 and Trilogy 200 devices identified in this letter may separate from the plastic backing to which it is attached. If this were to happen, the foam could potentially block air…
RecallHealth product recall | 2022-12-21
3T MRI Systems
Philips has identified an issue where a specific component failure in the gradient coil of where the affected 3T MR Systems identified above may act as a heat source, with a potential to produce smoke and/or fire. If the components failure…
RecallHealth product recall | 2022-12-16
Allergenic extract non pollens (Peanut allergy): False negative for skin test
Affected lot may result in a false negative for skin test (peanut allergy).
RecallHealth product recall | 2022-12-14
Evolut Pro+ Transcatheter Aortic Valve
Infolding is a known phenomenon and occurs when the valve frame folds inward along a vertical line away from the valve inflow and appears as a seam in the frame or as overlapping frame cells on radiographic imaging. Infolding is different and distinct…
RecallHealth product recall | 2022-12-13