Novum IQ Large Volume Pump
Contact the manufacturer if you require additional information.
|Lot or serial number
|Model or catalog number
|Novum IQ Large Volume Pump
Through ongoing internal testing, Baxter Corporation has observed isolated incident of over infusion on the Novum IQ LVP infusion pump during test runs at 0.1 ml/hr after 40 hours of continuous infusion within the 96-hour set change interval.
Baxter Corporation is issuing a follow-up communication to notify users that all of the sets on the Novum IQ LVP compatible sets list are acceptable to use at rates below 1mL/hr except for the product code 1W5000, which will be removed from the Novum IQ LVP list of compatible sets.
Recall start date: September 15, 2022
Previous recalls or alerts
Baxter Healthcare Corporation
1 Baxter Parkway, Deerfield, Illinois, United States, 60015