Karl Storz Injection Needles
Contact the manufacturer if you require additional information.
|Lot or serial number
|Model or catalog number
|Flexible Injection Needle
|26175 PD 810506 8598D
|Laryngeal Injection Needle
Validation testing of the efficacy of the manual cleaning for the injection needle was performed. The testing showed that the required efficacy level was not achieved. Different scenarios were tested, and all possible manual cleaning steps included, and the residual protein value remained above the acceptance criteria.
Recall start date: October 16, 2023
Karl Storz SE & Co. KG
Dr.-Karl-Storz Strasse 34, Tuttlingen, Germany, 78532