Please note: using the search filters may not provide accurate search results for records prior to November 2021. When searching for older records, please ensure that you use the include archive checkbox.
Displaying 1 - 4 of 4 items.
Dose IQ Safety Software
Baxter Corporation is issuing an urgent medical device correction due to a defect in the Dose IQ Safety Software used with Novum IQ Large Volume Pump, resulting in an invalid initial setting for the air-in-line threshold for any new care area created in…
RecallHealth product recall | 2023-06-07
Self-Righting Luer Slip And Lock Tip Caps
Baxter Corporation is issuing an urgent medical device recall for all non-expired lots of Self-Righting Luer Slip And Luer Lock Tip Caps listed due to the potential of the packaging seal not maintaining a sterile barrier for the tip caps. Recall start…
RecallHealth product recall | 2022-08-02
Prismax Control Unit
Baxter Corporation is issuing a correction for the Prismax. If the operator initiates therapy by entering a value other than the default value for the patient gain/loss limit or the return pressure drop limit and uses the same patient function, the…
RecallHealth product recall | 2022-03-29
Self-Righting Luer Slip Tip Cap (Discpac, 100 Pack), Yellow
Baxter Corporation has received one complaint of a customer getting self-righting luer lock tip caps, yellow (product code h938690025) mislabeled as self-righting luer slip tip caps, yellow (product code h93866100). Recall start date: Feb 28, 2022
RecallHealth product recall | 2022-03-14