Returning search results with filters:
Remove filter forBrand: Wright Medical Technology Inc. Clear all
Please note: using the search filters may not provide accurate search results for records prior to November 2021. When searching for older records, please ensure that you use the include archive checkbox.
Displaying 1 - 4 of 4 items.
Evolve Modular Radial Head System - Cocr Stem Component
Stryker has identified a nonconformance within two specific lots of evolve proline stems. Specifically, the outer label does not match the device inside the package. the affected lot of evolve proline stem 7.5mm +2 contains a size 9.5mm +4 stem, while…
RecallHealth product recall | 2022-08-31
Evolve Triad Plate Cutter
Stryker has identified a nonconformance relating to the Evolve Triad Plate Cutter provided in the Evolve Triad Instrument Kit (4951kit1). Specifically, the plate cutter is unable to meet sterilization requirements when it is sterilized in a double…
RecallHealth product recall | 2022-06-13
Infinity Resection Guide
Stryker has identified a nonconformance (via product complaint june 2021) relating to the infinity™ resection guide. specifically, the products were incorrectly color-coded with yellow ink instead of blue ink. The device is…
RecallHealth product recall | 2022-06-07
Charlotte Foot And Ankle System - MTP HEX Screws, Non-Sterile
Through internal investigations and product complaints, Stryker has become aware that two specific lots of the Charlotte Foot And Ankle System MTP HEX screws contain the incorrect product in the package. Wright Medical, the manufacturer of the Charlotte…
RecallHealth product recall | 2021-12-08