Ventana HE 600 System
Contact the manufacturer if you require additional information.
|Lot or serial number
|Model or catalogue number
|Ventana He 600 System
|More than 10 numbers, contact manufacturer.
This recall is regarding fluid leaks within the middle stainer module of the Ventana he 600 that likely resulted in electrical shorts at the plate heater
connections that then escalated into fires.
Recall start date: June 16, 2022
Ventana Medical Systems Inc
1910 E Innovation Park Drive, Tucson, Arizona, United States, 85755