Displaying 1 - 1 of 1 items.
Xvivo Organ Chamber
It has come to Xvivo knowledge that the sterile barrier (welded polybag) show defects in the weld for a number of batches of the Xvivo Organ Chamber. Defect was categorized as unsealed sections of the weld (compromised sterile barrier). To reinforce the…
RecallHealth product recall | 2022-04-25