Returning search results with filters:
Remove filter forAudience: Hospitals
Remove filter forRecall class: Type III
Remove filter forType: Recall Clear all
Please note: using the search filters may not provide accurate search results for records prior to November 2021. When searching for older records, please ensure that you use the include archive checkbox.
Displaying 1 - 8 of 8 items.
The pagewriter tc30/50/70 cardiograph device could potentially fail to charge while plugged into the ac power outlet. This is due to a faulty mainboard. based on a supplier analysis, the faulty mainboard was caused by…
RecallHealth product recall | 2022-05-30
Odan Itraconazole: Out of specification
The assay is out of specification in the affected lot
RecallHealth product recall | 2022-02-23
The esophageal/rectal/skin temperature probe's instructions for use, (IFU), contain inadequate instructions related to the cleaning and disinfection process for the reusable probes. The IFU instructs the user to clean the product with disinfecting agents…
RecallHealth product recall | 2022-02-21
LiquiBandFIX8® Open Hernia Mesh Fixation Device
During verification testing, a non-conformance was detected. Failure mode was uncontrolled leakage of glue from the distal tip of the Fix8 Open (FX002) device. Recall start date: Feb 3, 2022
RecallHealth product recall | 2022-02-11
Reddy-Dasatinib: out of specification
The dissolution is out of specification in the affected lot(s).
RecallHealth product recall | 2022-01-31
Nasacort Allergy 24H: Missing security seal
The affected lots are missing security seal on the external package.
RecallHealth product recall | 2021-12-22
CanPrev Omega Twist & Cyto Matrix Omega Boost: Labelling Issue
Affected lot(s) is/are labelled with incorrect expiry date and storage conditions found on the bottle.
RecallHealth product recall | 2021-12-01
VIDAS Protein C & TOXO IGG Avidity
Based on an unusual rate of complaints from the field for "Substrate error" leading to a potential delayed results on different References tested on all Systems, Biomerieux initiated an investigation. The substrate error…
RecallHealth product recall | 2021-11-05