Returning search results with filters:
Remove filter forAudience: Hospitals
Remove filter forIssue: Sterility Issue Clear all
Please note: using the search filters may not provide accurate search results for records prior to November 2021. When searching for older records, please ensure that you use the include archive checkbox.
Displaying 1 - 3 of 3 items.
Alliance Earloop Procedure Mask
Mold was found in lot Y591676 of the product. Two species of mold were identified that could pose a potential health risk to users. Recall start date: May 3, 2022
RecallHealth product recall | 2022-05-30
Pico50 Arterial Blood Sampler
Radiometer has been performing integrity testing of the Pico50 arterial blood sampler packaging material. Integrity studies included sterility testing to show that the integrity of the sterile barrier system is intact. In the 24 month real time study, 1…
RecallHealth product recall | 2022-04-25
Acufex Access Advanced Positioning Kits
Smith & Nephew, Inc., has initiated a field action to voluntarily remove three lots of the Acufex Access Advanced Positioning Kits due to the sterilization omission. The affected products were inadvertently shipped to the global distribution center…
RecallHealth product recall | 2021-12-21