Vantage Galan 3T
Contact the manufacturer if you require additional information.
|Lot or serial number
|Model or catalog number
|Vantage Galan 3T
It has been determined that the terminal block used to secure the power supply cable for the gradient coil of the MRI system was not secured as designed. Because of this, the cable became disconnected, and arcing occurred. Melting of the cable due to this arcing caused the cable securing parts below it to ignite, resulting in fire at the terminal block.
Recall Start Date: November 14, 2023
Canon Medical Systems Corporation
1385, Shimoishigami, Otawara-Shi, Tochigi-Ken, Japan, 324-8550