Vanta With Adaptivestim Technology
Contact the manufacturer if you require additional information.
|Lot or serial number
|Model or catalog number
|Vanta With Adaptivestim Technology
|More than 10 numbers, contact manufacturer.
Vanta Implantable Neurostimulator (INS) may malfunction after a cardioversion or defrbrillation procedure. The damage/malfunction to the device may no longer allow it to provide therapy or communicate with applications, resulting in a loss of stimulation and the need for an unanticipated surgery to explant/replace the INS.
Recall start date : June 6, 2023
710 Medtronic Parkway N.E., Minneapolis, Minnesota, United States, 55432