Aquac Uno H Reverse Osmosis System - Main Unit
Contact the manufacturer if you require additional information.
|Lot or serial number
|Model or catalogue number
|Aquac Uno H Reverse Osmosis System - Main Unit
|More than 10 numbers, contact manufacturer.
On June 14, 2022, the manufacturer Fresenius Medical Care Ag & Co. Kgaa notified Fresenius Medical Care Canada, Inc. that during regular quality inspections of Aquac Uno H devices, it was discovered that the internal ethernet cable had not been placed correctly during assembly. Instead of routing the cable outside the boundaries of the fuse board, it was routed across the board. Due to this, the cable may get in close contact to the fuse holders. This close contact can cause thermal damage to the outer jacket of the cable and apply the main voltage to the ethernet cable. If the corresponding ethernet socket outside of the device is touched, there is the risk of electrical shock to the user.
Recall start date: June 16, 2022
Fresenius Medical Care Ag & Co. Kgaa
Else-Kroner-Strasse 1, Bad Homburg, Germany, 61352